The NCD beamline, now NCD-SWEET, devoted to Small Angle and Wide Angle X-ray Scattering (SAXS, WAXS), is offering users further experimental possibilities and higher quality data.
The SAXS beamline of ALBA has gone through a major upgrade in 2017. Upgraded items in the SAXS WAXS experimental techniques (SWEET) involve a new monochromator system, a new photon counting detector (Pilatus 1M), a new sample table with an additional rotating stage, and a beam conditioning optics with µ-focus and GISAXS options.
The original double crystal monochromator (DCM) has been replaced by a channel-cut silicon (1 1 1), improving the beam stability at sample position up to 0.9% and 0.4% of the beam size horizontally and vertically, respectivelly.
Figure: Vertical beam profile with the Be lenses into the beam (Horizontal axis unit is mm). The plot is the derivative of an edge scan along the vertical direction. The horizontal beam profile shows a gaussian shape as well.